User:Elkinsearthworks/sandbox

Version

Change Implemented

3.0

Increase the user temperature max entry from 150 to 250 and decrease the min entry from -40 to -80 (degrees C and degrees F).

2.9

Remove sensor warm-up timer during calibration if v1.08 (sensors always on) firmware is on Envision.

Configuration editor - allow loading of software to mobile device.

Don't allow commas in the add new technician field. Causes the program to hang when first booting or selecting a site.

Change minimum pump time from 60 seconds to 30 seconds in configuration editor.

Fix view previous data screen. Was displaying zero for GPS coordinates and temperature.

Don't automatically lock adjusted on pressures screen when advancing to the next screen.

Don't allow a pressure sensor cal/zero after locking gas concentration values to cause values to revert to zero before being stored.

2.8

Fix issue of not logging meter ID and verify GPS coordinates for probes with encryption option.

2.7

Revert to compact framework v2.0 to keep program from crashing when Bluetooth is dropped.

2.6

Allow editing multiple rows at the same time in configuration editor.

Allow import of GEM ID sets in configuration editor.

Allow less than 1.0% span gas to be used.

Configuration editor - check to make sure "File -> Open" is an Envision file.

Fix problem of missing line in calibration.xml file causing Bluetooth to drop immediately after connecting.

2.5

Update software to .Net v3.5 CF

Fix flow entry issue. When ID is set to user input, must hit backspace key to delete "NA" and remembers previous data but should not.

Add decimal precision to user options screen.

Reverse order of CO2 and CH4 on gas cal screen. Calibrating CO2 first gives slightly better accuracy due to methane sensor cross-sensitivity.

Fix problem of pressures starting at a very high number when screen first displayed.

2.4

Display "Service Unit Soon" message if within 30 days of annual calibration due.

Make "Lock Adjusted" buttons change colors when pressed on pressure measurement screen.

Allow manual temperature input even when thermistor is connected.

Fix program crash if cal timer times out and no Envision connected.

Don't allow the slash character when creating new site names in configuration editor.

Add "comments" right soft key to "analyze" screen when doing probes.

On probe pressures screen change right soft key to "next id" and add button for "analyze".

2.3

Internal only.

2.2

Add option/button to change degF to degC to user options screen.

Compensate CH4 measurement for CO2 cross sensitivity.

Don't allow program to crash after having measured/stored a probe, returning to the same probe, and pressing "view previous data" button.

Give warning message if hitting "Restore Factory Calibration" button will render unit unusable.

Turn wifi off automatically when program launches to keep from interfering with Bluetooth.

Show calibration due message if Envision cal date is older than one year.

Automatically reset calibration values in Nomad if unit was recently factory calibrated.

Add message saying factory cal has been restored when hitting restore factory cal button.

Fix configuration editor. If file imported, save operation results in exception error. Also, when file imported, not all well types show up when editing an ID.

Fix configuration editor. Don't allow error if selecting a mobile device file from file pull down.

Compensate gas measurements for pressure changes.

Show "service unit soon" message if sensor has drifted outside of acceptable tolerance window during calibration.

Log pressure sensor zero offsets to calibration file.

Require license key to run main program.