User:KJenkin94/sandbox

Fast atom bombardment (FAB) is an ionization technique used in mass spectrometry in which a beam of high energy atoms strikes a surface to create ions. It was developed by Michael Barber at the University of Manchester in 1980. When a beam of high energy ions is used instead of atoms (as in secondary ion mass spectrometry), the method is known as liquid secondary ion mass spectrometry (LSIMS). In FAB and LSIMS, the material to be analyzed is mixed with a non-volatile chemical protection environment, called a matrix, and is bombarded under vacuum with a high energy (4000 to 10,000 electron volts) beam of atoms. The atoms are typically from an inert gas such as argon or xenon. Common matrices include glycerol, thioglycerol, 3-nitrobenzyl alcohol (3-NBA), 18-crown-6 ether, 2-nitrophenyloctyl ether, sulfolane, diethanolamine, and triethanolamine. This technique is similar to secondary ion mass spectrometry and plasma desorption mass spectrometry.

Ionization mechanism
FAB is a relatively low fragmentation (soft) ionization technique and produces primarily intact protonated molecules denoted as [M + H]+ and deprotonated molecules such as [M - H]−. Radical cations can also be observed in a FAB spectrum in rare cases. FAB was designed as an improved version of SIMS that allowed for the primary beam to no longer cause damaging effects to the sample. The major difference between the two techniques is the difference in the nature of the primary beam used. For LSIMS, Cesium, Cs+ ions make up the primary beam and for FAB the primary beam is made up of Xe or Ar atoms. Xe atoms are used because they tend to be more sensitive than Argon atoms due to their larger masses and more momentum. For the molecules to be ionized by FAB, first the slow moving atoms (Xe or Ar) are ionized by colliding electrons. Those slow moving atoms are then ionized and accelerated to a certain potential where they develop into fast moving ions that become neutral in a dense cloud of excess natural gas atoms that make a flowing stream of high translational energy atoms. Although the exact mechanism of how the samples are ionized have not been fully been discovered, the nature of its ionization mechanism is similar to matrix-assisted laser desorption/ionization (MALDI) and chemical ionization.

Matrices and Sample Introduction
As previously stated, in FAB the samples are mixed with a non-volatile environment (matrix) in order to be analyzed. FAB uses a liquid matrix that is mixed with the sample in order to provide a sample ion current that is sustained, reduce damages made to the sample by absorbing the impact of the primary beam, and keeps the sample molecules form aggregating. The liquid matrix, like any other matrix, most importantly provides a medium that promotes sample ionization. The most widely accepted matrix for this type of ionization is glycerol. Choosing the appropriate matrix for the sample is crucial because the matrix can also influence the degree of fragmentation of the sample (analyte) ions. Once the sample and matrix are mixed together properly, the sample can then be introduced to FAB analysis. The normal method of introducing the sample-matrix mixture is through a insertion probe. The sample-matrix mixture is loaded on a stainless steel sample target on the probe, which is then placed in the ion source via a vacuum lock. The alternative method of introducing the sample is by using a device called continuous flow fast atom bombardment (CF)-FAB.

Continuous flow fast atom bombardment
In continuous flow fast atom bombardment (CF-FAB), the sample is introduced into the mass spectrometer insertion probe through a small diameter capillary. (CF)-FAB was developed to minimize the problem of poor detection sensitivity that is caused by an excess of the matrix background that results in a high matrix-to-sample ratio. When a metal frit is used to disperse the liquid on the probe, the technique is known as frit FAB. Samples can be introduced by flow injection, microdialysis, or by coupling with liquid chromatography. Flow rates are typically between 1 and 20 µL/min. CF-FAB has a higher sensitivity compared to static FAB

Applications
The first example of the practical application of this FAB was the elucidation of the amino acid sequence of the oligopeptide efrapeptin D. This contained a variety of very unusual amino acid residues. The sequence was shown to be: N-acetyl-L-pip-AIB-L-pip-AIB-AIB-L-leu-beta-ala-gly-AIB-AIB-L-pip-AIB-gly-L-leu-L-iva-AIB-X. PIP = pipecolic acid, AIB = alpha-amino-isobutyric acid, leu = leucine, iva = isovaline, gly = glycine. This is a potent inhibitor of mitochodrial ATPase activity. Another application of FAB includes its original use for the analysis of condensed-phase samples. FAB can be use for measurements of the molecular weight of samples below 5000 Da, as well as their structural characteristics. FAB can be paired with various mass spectrometers for data analysis, such as with a quadrupole mass analyzer, liquid chromatography–mass spectrometry, and more.