User talk:Epowersunny

XB-1000 Infusion Pump and syringe pump, please contact Ms. Sunny for details. XB-1000 infusion pump CHARACTERISTICS: 1&gt; 2 control ways: Can be used in either drop counting control way or volumetric control way ml/h or drops/min 2&gt; High speed: 1ml/h~1000ml/h or 1 drop/min ~250 drop/min. 3&gt; 3 levels of occlusion value: High, middle and low. 4&gt; Hard based free-flow protection: Avoid any free flow or leakage, prevent any danger to patients. 5&gt; Drops detector: Can detect the drops infused, automatically detect the IV line type and ensure the accuracy for any IV line. 6&gt; Applicable IV Sets: Any standard IV sets. 7&gt; Rechargeable internal battery. 8&gt; Visible and audible alarm: When any malfunction, error, air-on-line, occlusion, error flow rate and so on. 9&gt; KVO function: After infusion is complete, the pump automatically switches to the smallest flow to prevent creating the thrombus. 10&gt; Bolus function: When during infusion, press a key to infuse at the rate of 400ml/h.

SPECIFICATIONS: 1&gt; Accuracy: +/-3% ~5% 2&gt; Delivery Rate Range: 1~250 drop/min(1drop/min step) or 1ml/h~1000ml/h 3&gt; Volume Limit Range: 0~9999ml (1 ml step) 4&gt; Bolus rate: 500ml/h 5&gt; KVO Rate: 1ml/h~5ml/h 6&gt; Alarms: Air-in-line, Infusion complete(a volume limit has been set), Empty container, Occlusion, Door open, Over time, No drop sensor, Operation error, Low battery and other device failures. 7&gt; Power supply: AC100~ 240V+/-10%, 50/60Hz; DC 12V 8&gt; Power Consumption: &le; 28VA 9&gt; Shock Protection: Class I, Type BF 10&gt; Dimensions: 135mm*120mm*245mm 11&gt; Net Weight: &lt; 2.4 kg 12&gt; Operating Condition: Ambient temperature: 5 to 40C Relative humidity: 20 to 90% Storage Condition: Ambient temperature: -10 to 55C Relative humidity: 10 to 90% 13&gt; Accessories: AC power line, Operation manual