Cold finger



A cold finger is a piece of laboratory equipment that is used to generate a localized cold surface. It is named for its resemblance to a finger and is a type of cold trap. The device usually consists of a chamber that a coolant fluid (cold tap water, or perhaps something colder) can enter and leave. Another version involves filling the device with a cold material (examples: ice, dry ice or a mixture such as dry ice/acetone or ice/water).

Typically a cold finger is used in a sublimation apparatus, or can be used as a compact version of a condenser in either reflux reaction or distillation apparatus. Many commercially available rotary evaporators can be purchased with a cold finger in place of a Dimroth condenser, for example. When used as a condenser in a rotary evaporator, cold fingers can be cooled to a lower temperature of &minus;78 °C (dry ice), compared with water condensers that can be cooled to &minus;40 °C (ethylene glycol/water mixture). The lower temperature achieved reduces the quantity of volatile material exhausted into the air.